Информация по оборудованию
|AVO CT38 Аналоговый мультиметр|
I. This unit is a multi-range electronic measuring instrument designed to eliminate the necessity for the provision of a wide range of separate instruments normally required for the testing and setting-up of radio, radar and electronic equipment. It provides 97 ranges of measurement covering DC and AC current and voltage, resistance, and power output and has been designed for use under pan-climatic conditions. The accuracy on DC ranges is ^ 2 per cent FSD and on AC voltage ranges it is ± 3 per cent FSD measured at 50 c/s. 2. The instrument is AC mains operated and may be used on any supply system of from 105 to 125 volts, 45 to 66 c/s and 195 to 255 volts, 45 to 66 c/s. Connection to the mains supply is made bv a mains lead and connector which is provided with the instrument. 3. The mechanical construction of the unit has been designed to make servicing as simple as possible. The front panel is of aluminium alloy and bolted to it is a light, braced aluminium alloy framework of rectangular form which carries all the components of the instrument with the exception of those such as switches and the meter which are carried by the front panel. The majority of smaller components are arranged on tag-boards, and the construction is such that these tag-boards are easilv removable for servicing or inspection. All wiring is made up into cable-forms. 4. The overall size is 16 in. wide, 10 in. high and l2iin. deep when the unit is enclosed in its metal case. For protection in transit, a deep metal lid is fitted over the front of the unit, held in position by four screws, and in this lid are stowed the various leads and connectors of the instrument. These include the mains cable, the RF measuring head unit extension lead, high-voltage multiplier and multi-range shunt resistors, spare fuses, pilot lamp, and millivoltmeter valve.
Эти руководства доступны для вышеуказанного оборудования:
|AVO -- CT38 -- Руководство по техническому обслуживанию|
|Ручной тип:||Руководство по техническому обслуживанию|
|Размер:||2.58 Mbytes (2707572 Bytes)|
|Качество:||Сканированный документ, читая частично плохо, частично не читается.|
|Дата загрузки:||07 Март 2015|
|Загрузки:||251 начиная с 07 Март 2015|
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.