Informações do equipamento
|Philips PM2422A Multímetro digital|
|Categoria:||Instrumentos de medição|
The Philips Digital multimeter PM 2422 A is an accurate 3+1 digit instrument with a maximum reading of 1999 to measure: - DC and AC voltages of 100 pV ... 1000 V (600 V AC,) 5 ranges - DC and AC currents of 100 nA ... 2 A, 5 ranges - Resistances of 0.1 Q ... 20 Mfi, 6 ranges - DC high voltages up to 30 kV with the EHT Probe PM 9246 - UHF voltages with UHF Probe PM 9200 up to 800 MHz. For frequencies above 30 MHz it is advisable to use T-connector PM 9250 in conjunction with this Probe. - DC and AC currents up to 31.6 A with shunt PM 9244 - AC currents up to 100 A with current transformer PM 9245 Operation of the instrument is easy. Selection of the measuring ranges .$ effected by a rotary switch, and AC and DC selection by push-buttons. The decimal point is set automatically by switching the ranges. The polarity of direct voltages and currents is indicated automatically. Only the -sign is displayed. The instrument is protected against overloads on all ranges up to at least 250 V. This efficiently designed instrument can be employed for all applications in modern electronics. TECHNICAL DATA Properties expressed in numerical values with tolerances are guaranteed by the manufacturer. Numerical values without tolerances are given for general information and represent the properties of an average instrument. The data below are applicable at 23 C + 1 C unless otherwise stated.
Estes manuais estão disponíveis para os equipamentos acima:
|Philips -- PM2422A -- Serviço e Manual do Usuário|
|Tipo manual:||Serviço e Manual do Usuário|
|Tamanho:||26.91 Mbytes (28212956 Bytes)|
|Manual-ID/número:||9499 470 08802|
|Data:||01 Setembro 1974|
|Qualidade:||Scaned documento, todas legível.|
|Upload de data:||29 Janeiro 2017|
|Downloads:||240 desde 29 Janeiro 2017|
GENERAL / ALLGEMEINES / GENERALITES 1. INTRODUCTION 6 EINLEITUNG 20 INTRODUCTION 34 II. TECHNICAL DATA 6 TECHNISCHE DATEN 20 CARACTERISTIQUES TECHNIQUES 34 III. ACCESSORIES 11 ZUBEHOER 25 ACCESSOIRES 39 IV. PRINCIPLE OF OPERATION 13 WIRKUNGSWEISE 27 PRINCIPE DE FONCTIONNEMENT 41 DIRECTIONS FOR USE / GEBRAUCHSANWEISUNG / MODE D'EMPLOI V. INSTALLATION 16 INSTALLATION 30 MISE EN PLACE 48 VI. OPERATION 16 BEDIENUNG 30 MISE EN OEUVRE 48 SERVICE DATA VII. CIRCUIT DESCRIPTION VIII. ACCESS IX. MAINTENANCE AND SERVICE X. CHECKING AND ADJUSTING SURVEY XI. CHECKING AND ADJUSTING XII. PARTS LIST
|Iniciar Download grátis|
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.