Informações do equipamento
|Wavetek 751A Outro|
|Categoria:||Instrumentos de medição|
The Models 751 A, 752A and 753A are programmable Brickwall™ Filters offering near-ideal passband and stopband characteristics. The instruments employ 7th-order (7 poles and 6 zeroes) Elliptic or Cauer filters with design values of0.3dBpeak- to-peak passband ripple and 85 dB stopband attenuation. The transition from passband to stopband has a 115 dB/octave rolloff rate. For low pass response, the stopband is at 1.7 x cutoff frequency; high pass response, the stopband is 0.6 x cutoff for frequency. The cutoff frequency may be programmed from 10 Hz to 99 kHz with 2-digit resolution and 1 Hz to 9 Hz with 1-digit resolution. Pre-filter, and (where available) postfilter gain may be programmed in 10 dB steps, up to the maximum available. These instruments are extremely valuable in signal- processing applications for band-limiting prior to sampling to prevent aliasing, or for separating a narrow spectrum of interest from noise or other interfering signals. In addition to the extremely Sharp filtering, they provide programmable low noise amplification for applications involving low-level signals and overload indicators to ensure operation within their linear dynamic range. The basic differences of the three models are summarized here. Model 751A is a single-channel band pass filter which consists of a high-pass section followed by a low- pass section. Low-pass and high-pass cutoff frequencies may be independently programmed. Prefilter gain, up to 40 dB, may be programmed, as well as postfilter gain, up to 20 dB. In addition, there are individual input and output overload indicators.
Estes manuais estão disponíveis para os equipamentos acima:
|Wavetek -- 751A -- Serviço e Manual do Usuário|
|Tipo manual:||Serviço e Manual do Usuário|
|Tamanho:||13.80 Mbytes (14473940 Bytes)|
|Qualidade:||Scaned documento, todas legível.|
|Upload de data:||17 Novembro 2014|
|Downloads:||1090 desde 17 Novembro 2014|
GENERAL DESCRIPTION 1.1 INTRODUCTION ... 1.1.1 Model 751A ... 1.1.2 Model 752A ... 1.1.3 Model 753A ... 1.2 OPTIONS ... 1.3 DEFINITIONS ... 1.4 SPECIFICATIONS ... 1.4.1 Cutoff Frequency ... 1.4.2 Gain (Pre-filter) ... 1.4.3 Filter Characteristics ... 1.4.4 Input Characteristics ... 1.4.5 Output Characteristics ... 1.4.6 General ... 1.4.7 Options ... INITIAL PREPARATION 2.1 INSPECTION ... 2.2 MECHANICAL INSTALLATION ... 2.3 ELECTRICAL INSTALLATION ... 2.3.1 Power Connection ... 2.3.2 Signal Connection ... 2.3.3 GPIB Option ... 2.3.4 BCD Option ... OPERATION 3.1 INTRODUCTION ... 3.2 CONTROLS AND INDICATORS ... 3.3 TYPICAL OPERATING PROCEDURES ... 3.3.1 Model 751A ... 3.3.2 Model 752A ... 3.3.3 Model 753A ... 3.4 REMOTE PROGRAMMING: BCD Interface 3.5 REMOTE PROGRAMMING: GPIB Interface 3.5.1 Introduction ... 3.5.2 Device Address Code ... 3.5.3 Initialization ... 3.5.4 Listen State ... 3.5.5 Sending Data ... CIRCUIT DESCRIPTION 4.1 INTRODUCTION ... 4.2 MODEL 751A ... 4.3 MODEL 752A ... 4.4 MODEL 753A ... 4.5 ELLIPTIC FILTERS ... CALIBRATION 5.1 FACTORY REPAIR ... 5-1 5.2 CALIBRATION ... 5-1 SECTION 6 TROUBLESHOOTING 6.1 FACTORY REPAIR ... 6-1 6.2 TROUBLESHOOTING ... 6-1 6.2.1 Filter Troubleshooting ... 6-1 6.2.2 Optional GPIB Interface Troubleshooting ... 6-1 SECTION 7 PARTS AND SCHEMATICS 7.1 DRAWINGS ... 7-1 7.2 ADDENDA ... 7-1 7.3 ORDERING PARTS ... 7-1
|Iniciar Download grátis|
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.