|Advantest Corporation R3271 Versterker|
|Categorie:||Audio, Video, TV, Multimedia|
Outline of the Analyzer The R3265/3271 series spectrum analyzer which employs synthesized iocal oscillation assures"? high stability in spectrum analysis. Frequency range : input range - 140dBm to +30dBm (R3265) 100Hz to 8.0 GHz (R3265) ,100Hz to 26.5GHz (R3271) : - T ^ d B m to + 30dBm (R3271 ) Display range : 95dB In the wide range as described above, the analyzer features the following basic performances! maximum resolution 10Hz, residual FM ЗНр.р and noise sideband -112dBc/Hz (at 10kHz fror the carrier). !t is equipped with full remote control GPiB and memory card function fo| saving/recalling waveform data and panel setting conditions. Features of the Analyzer Ф The analyzer perm its sweeping over a wide frequency range at once: from 100Hz tel 26.5GHz (R3271); from 100Hz to 8GHz (R3265). Log sweep is also enabled in the rangF from 1kHz to 1GHz................................. © High frequency resolution of 10Hz at maximum permits analysis of adjacent signals апф spurious at high frequency. ® High-precision frequency measurement The built-in reference crystal oscillator with aging rate of 2 x 10_ s/day permits to measure^ very week signals such that cannot be measured by any counter, at a resolution of 1 Hz in:, the counter mode of frequency measurement. @ © Setting conditions and waveform data can be stored by using memory card. The electric field strength can be viewed and read directly after compensating the antenna calibration coefficient, and the QP value based on the CISPR specification can be: observed directly. © Various enhanced functions supported by digital indications The information required for spectrum analysis is entirely displayed on the CRT together, with signal traces. The digital memory screen realizes fHckerless display. Various marker; functions assures accurate and easy reading even in the manual mode. Ф Sweep time of 50/ys can be set in the ZERO SPAN mode, which facilitates analysis of thè: burst wave on the time axis. Two independent channels of digital memory enable simultaneous display of two screens, <§) A full-remote GPiB is used as a powerful measurement system component.
Deze handboeken zijn beschikbaar voor de bovengenoemde apparatuur:
|Advantest Corporation -- R3271 -- Gebruikershandleiding|
|Manueel de Type:||Gebruikershandleiding|
|Grootte:||15.30 Mbytes (16043969 Bytes)|
|Datum:||22 maart 1991|
|Kwaliteit:||Gescande document, lezen deels slecht, deels niet leesbaar.|
|Uploaden datum:||17 augustus 2016|
|Downloads:||363 Sinds 17 augustus 2016|
TABLE OF CONTENTS 1. GENERAL 2. PANELS 3. BASIC OPERATIONS 4. EXAMPLES OF MEASUREMENT 5. EXPLANATION OF FUNCTIONS 6. GPIB:REMOTE PROGRAMMING 7. INSPECTION 8. OPERATION DESCRIPTION 9. SPECIFICATIONS APPENDIX
|Start gratis downloaden|
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.