|Hewlett Packard 8754A Network Analyzer|
The HP Model 8754A Network Analyzer consists of an internally leveled, 4- to 1300-MHz, swept source. Polar or rectangular measurements are displayed on a cathode-ray tube (CRT) through two independent channels. With appropriate signal-routing accessories, the 8754A is a fully integrated stimulus/response test system that measures magnitude and phase characteristics of linear networks by comparing the incident signal with the signal transmitted by the device under test or reflected from its input. 1-20. The basic transmission measurements are insertion loss or gain, insertion phase, and transmission coefficient (S2i or Si2)· The basic reflection measurements are return loss, from which SWR can be calculated, and reflection coefficient (Sjj or S22X from which impedance can be calculated or read from a Smith chart overlay.
Deze handboeken zijn beschikbaar voor de bovengenoemde apparatuur:
|Hewlett Packard -- 8754A -- Service-en gebruikershandleiding|
|Manueel de Type:||Service-en gebruikershandleiding|
|Grootte:||26.60 Mbytes (27888395 Bytes)|
|Kwaliteit:||Gescand document alle leesbaar.|
|Uploaden datum:||30 mei 2015|
|Downloads:||351 Sinds 30 mei 2015|
This manual applies directly to instruments whose serial number prefix is 1914A. This Operating and Service Manual contains information required to install, operate, test, adjust, and service the Hewlett-Packard Model 8754A Network Analyzer. Figure 1-1 shows the instrument and Accessories supplied. This section covers instrument identification, description, options, accessories, specifications, and other basic information. 1-3. This manual is divided into eight sections, which provide the following information: a. SECTION I, GENERAL INFORMATION, presents a brief description of the instrument, specifications, and lists of Accessories and recommended test equipment. b. SECTION II, INSTALLATION AND OPERATION VERIFICATION, presents information relative to initial inspection, preparation for use, mounting, packing, shipping, and operation verification. c. SECTION III, OPERATION, presents instructions for operation of the instrument. d. SECTION IV, PERFORMANCE TESTS, presents procedures required to verify that performance of the instrument is in accordance with published specifications. e. SECTION V, ADJUSTMENTS, presents procedures required to properly adjust and align the instrument after repair. f. SECTION VI, REPLACEABLE PARTS, presents information required to order all parts and assemblies. g. SECTION VII, MANUAL BACKDATING CHANGES, presents backdating information to make this manual compatible with earlier shipment configurations. h. SECTION VIII, SERVICE, presents circuit descriptions, schematic diagrams, component location diagrams, and troubleshooting procedures to aid the user in maintaining the equipment.
|Start gratis downloaden|
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.