Información sobre equipo
|Valhalla Scientifiс Inc . 2100 Analyzer|
Valhalla Scientifiс Inc .
|Categoría:||Los instrumentos de medición|
|Descripción:||Digital Power Analyzer|
Valhalla Scientific Models 2100 and 2101 Digital Wattmeters are accurate, low-cost In struments to aid engineering, production test and quality assurance departments in the determina tion of product power consumption from DC and AC power lines. The instruments feature dual, in dependent digital displays. One display provides a continuous indication of True Power in Watts. The other display is sw itch selectable between Amperes (True RMS) and Volts (True RMS). 1-3 The Model 2100 and Model 2101 provide a fast and convenient method of determining product ef ficiency, power factor and true RMS current. Phase angle relationships may be calculated through utilization of the displayed digital data. 1-4 The Digital Wattmeters use a unique four- quadrant complex waveform analog m ultiplier to derive a DC signal equivalent to the integrated pro duct of the instantaneous value of line voltage and current. The resultant wattage value is converted for presentation on a 4-1/2 digit display. Power ranging provides maximum resolution from 10 m illiwatts to 12000 watts on the Model 2100 and from 1 m illiw att to 6000 watts on the Model 2101. True RMS current from 100 microamperes (max resolution) through 19.99 amperes with a crest fac tor of 2.5:1 may be monitored by selecting the cur rent display with the front panel switch. True RMS voltage from 7.5 volts to 600 volts may be monitored on the Model 2100 by selecting the voltage display. On the 2101, the range is from 1.5 volts to 300 volts. Four binding posts on the rear panel provide convenient means of making the measured line and load connections. 1-5 The Digital Wattmeters are designed to operate from selectable 115/230V AC power. The loads may be operated from power sources up to 600 volts on the Model 2100 and up to 300 volts on the Model 2101. These sources may be indepen dent of instrument power and may be DC, or AC from 40 Hz to 20 KHz. Peak voltage limitation is 1500 volts for the Model 2100 and 750 volts for the Model 2101. There are three selectable current display ranges identified on the front panel as .2,2 and 20 amperes. DC current may be displayed up to 0.5000, 5.000 and 20.00 amperes. AC sinewave current may be displayed up to 0.3500, 3.500 and 20.00 amperes. Voltage display ranges are also selectable in three steps; 150.00, 300.0 and 600.0 on the Model 2100 and 30.00, 150.00 and 600.0 on the Model 2101
Estos manuales están disponibles para el equipo anterior:
|Valhalla Scientifiс Inc . -- 2100 -- Manual del usuario|
|Tipo de Manual::||Manual del usuario|
|Tamaño::||228.58 Kbytes (234064 Bytes)|
|Número de identificación:|
|Fecha:||25 marzo 1988|
|Calidad:||Documento Scaned, todo legible.|
|Añadir fecha:||10 abril 2017|
|Descargas:||542 desde 10 abril 2017|
Only a few pages!
|Descarga gratuita de inicio|
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.