|Philips PM 5190 Generator|
|Description:||LF Synthesizer 1mHz-2MHz|
The LF Synthesizer PM 5190 is a versatile high performance signal Generator. It produces three wave forms, namely sine wave, square wave or triangular wave form in the frequency range from 1mHz to 2MHz. The instrument is designed for most professional applications, where a stable signal source is asked for, for development laboratories, including calibration purposes and advanced educational programs. The frequency settings of the Synthesizer are highly accurate due to X-tal controlled generation. All basic signal parameters, i.e. frequency, amplitude, dc offset and wave form are set via keyboard in the local mode of operation. In addition all these parameters are remotely programmable via the built-in IEC bus Interface. In local as well as in remote operational mode these parameters are indicated by a bright seven segment LED display in combination with dimension and wave form LED1s. A microprocessor represents the central data processing and Control Unit of the instrument. It accepts input data from the keyboard or via IEC bus. After diagnosting and processing these data it stimulates the various functional units of the instrument for proper response via assigned I/O ports. In case of incompatible settings the microprocessor initiates the LED display flashing. A rub-out facility allows fast alteration or correction of keyboard input data. The inherent signal generation technique of the instrument is called direct digital signal synthesis: Succeeding digital samples of the signal are computated by a fast processing TTL circuitry and converted to analogue. This technique effects extreme fast responding on alternate frequency settings. Remote data programming is organized according to the international standard “Interface System for Programmable Measuring Apparatus", called IEC bus. So via the built-in IEC bus Interface the PM 5190 cooperates with all Philips and non-Philips IEC bus instruments. The dc output voltage may be selected separately or whenever used as dc offset voltage added to the output signal. The Generator provides external amplitude modulation facility. For TTL circuitry testing and stimulating a separate TTL output is available. All these facilities are built into a compact portable housing* The clearly designed frontpanel and the simplicity of keyboard input as well serve for convenient operating the instrument.
These manuals are available for the above equipment:
|Philips -- PM 5190 -- Service and User Manual|
|Manual Type:||Service and User Manual|
|Size:||37.66 Mbytes (39491839 Bytes)|
|Language:||english, german, french|
|Manual-ID/Number:||9499 453 00502 and/or 9445 051 90001|
|Date:||15 May 1980|
|Quality:||Scanned document, all readable.|
|Upload date:||01 August 2020|
|Downloads:||12 since 01 August 2020|
1• GENERAL 1.1. Introduction E 1- 1.2. Technical data E 1-: 1.3. Accessories E 1-1 1.4. Operating principle E 1-< 2. INSTALLATION 2.1. Safety regulations E 2- 2.2. Mounting E 2-: 2.3. Earthing E 2-: 2.4. Dismantling the instrument E 2-; 2.5. Mains connection e 2-: 3. OPERATING INSTRUCTIONS 3. 1. Controls and sockets E 3- 3.2. Operation E 3-i 3.3. IEC bus E 3-' 3.3.1. Introduction E 3-< 3.3.2. Remote operation E 3- 3.3.3. Programming examples E 3- 3.3.4. Application examples E 3-; SERVICE PART 4. CIRCUIT DESCRIPTION 4. 1. Oscillator 4-1 4.2. IEC bus interface 4-1 4.3. Control circuit 4-1 4.4. Keyboard and display 4-2 4.5. Digital signal synthesis 4-2 4.6. Digital to analog converter 4-5 4.7. Low-pass filter 4-5 4.8. Square wave generator 4-5 4.9. Square wave pre-amplifier 4-6 4. 10. Amplitude modulation 4-6 4.11. Buffer amplifier, amplitude control/ compensation 4-6 4. 12. DC generator 4-7 4.13. Output amplifier, attenuator 4-7 4. 14. Power supply 4-7 5. ACCESS TO PARTS 5.1. Cabinet 5-1 5.2. Pushbuttons 5-1 5.3. Textplate 5-1 5.4. Unit 1 5-1 5.5. Unit 2 5-1 5.6. Unit 3/ Unit 4 5-2 5.7. Fuse, mains transformer 5-2 6. CHECK AND ADJUSTMENT 6.1. General 6-1 6.2. Overall functional test 6-1 6.3. Diagnostic program 6-1 6.4. Adjustments 6-5 6.5. Final check 6-9 6.6. Check after repair and maintenance 6-10 7. PARTS LIST 7-1 7.1. Mechanical parts, miscellaneous, parts not on unit 7-1 7.2. Electrical parts 7-2
|Start Free Download|
This policy contains information about your privacy. By posting, you are declaring that you understand this policy:
- Your name, rating, website address, town, country, state and comment will be publicly displayed if entered.
- Aside from the data entered into these form fields, other stored data about your comment will include:
- Your IP address (not displayed)
- The time/date of your submission (displayed)
- Your email address will not be shared. It is collected for only two reasons:
- Administrative purposes, should a need to contact you arise.
- To inform you of new comments, should you subscribe to receive notifications.
- A cookie may be set on your computer. This is used to remember your inputs. It will expire by itself.
This policy is subject to change at any time and without notice.
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.