|Kenwood CS-1010 Oscilloscope|
Vertical axis has high sensitivity and wide bandwidth and especially covers fully specified frequency response at 2mV/div for all models. CS-1022, 1021 ; Dual Trace 1 mV/div : DC to 10 MHz, -3 dB 2mV/div : DC to 20 MHz, -3 dB CS-1012 ; Dual Trace 1 mV/div : DC to 7 MHz, -3 dB 2mV/div : DC to 10 MHz, -3 dB CS-1020 ; Single Trace 1 mV/div : DC to 10 MHz, -3 dB 2mV/div : DC to 20 MHz, -3 dB CS-1010 ; Single Trace 1 mV/div : DC to 7 MHz, -3 dB 2mV/div : DC to 10 MHz, -3 dB • Vertical sensitivity range is selectable from 1 mV/div to 5V/div with rotary switch continuously. • Time base permits the high sweep speed. CS-1022, 1020; 20nsec/div (x10 MAG) CS-1021, 1012, 1010; 50nsec/div (x10 MAG) • Vertical sensitivity error and sweep rate error are ±3% and accurate measurements are provided. • The 150 mm rectangular CRT with internal graticule provides high brightness and accurate measurements, free of parallax error. CS-1022, 1020; domed mesh type CRT with post deflection acceleration and high brightness phosphors (acceleration voltage; 6 kV). CS-1021, 1012, 1010; high brightness CRT (acceleration voltage; 2 kV). • For convenience in making rise time measurements, the 0%, 10%, 90% and 100% levels are marked on the graticule scale of the CRT. • Trace rotation is electrically adjustable from the front panel. • By SCALE ILLUM control, the waveform is easy observed in the dark and the photograph of the waveform is easy provided. (Except CS-1021) • Selectable AUTO FREE RUN function provides sweep without trigger input signal. • The FRAME-LINE switch provides selection of sync pulse for sweep triggering from small amplitude to large amplitude without adjusting when viewing composite video waveforms. • For CS-1022, 1021 and 1012, vertical mode automatically provides the trigger signal with TRIG SOURCE and V. MODE switches. • X-Y operation is easy provided by one-touche. CS-1022, 1021, 1012; CH1 Y axis, CH2 X axis CS-1020, 1010; VERT INPUT Y axis EXT TRIG INPUT X axis • CS-1022 and 1012 are provided with CH1 OUTPUT terminal to monitor input signal of CH1. • CS-1020 and 1010 are provided with VERT OUTPUT terminal to monitor input signal of VERT INPUT.
These manuals are available for the above equipment:
|Kenwood -- CS-1010 -- User Manual|
|Manual Type:||User Manual|
|Size:||5.87 Mbytes (6160360 Bytes)|
|Date:||01 December 1991|
|Quality:||Scanned document, all readable.|
|Upload date:||17 May 2020|
|Downloads:||8 since 17 May 2020|
SAFETY ... FEATURES ... SPECIFICATIONS ... PREPARATION FOR USE ... CONTROLS AND INDICATORS ... FRONT PANEL ... REAR PANEL ... OPERATION ... INITIAL STARTING PROCEDURE ... (1) NORMAL SWEEP DISPLAY OPERATION ... (2) MAGNIFIED SWEEP OPERATION ... (3) X-Y OPERATION ... (4) VIDEO SIGNAL OBSERVATION ... APPLICATION ... Probe COMPENSATION ... TRACE ROTATION COMPENSATION ... DC VOLTAGE MEASUREMENTS ... MEASUREMENTS OF THE VOLTAGE BETWEEN TWO POINTS ON A WAVEFORM ... ELIMINATION OF UNDESIRED SIGNAL COMPONENTS ... TIME MEASUREMENTS ... FREQUENCY MEASUREMENTS ... PULSE WIDTH MEASUREMENTS ... PULSE RISE TIME AND FALLTIME MEASUREMENTS ... TIME DIFFERENCE MEASUREMENTS ... PHASE DIFFERENCE MEASUREMENTS ... RELATIVE MEASUREMENTS ... APPLICATION OF X-Y OPERATION ... ACCESSORIES ...
|Start Free Download|
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.