|Hewlett Packard 113BR Frequency counter|
|Description:||Frequency Divider and Clock|
The Model 113BR is a component instrument in the Hewlett-Packard frequency and time standard system (see paragraph 1-8). Instrument specifications are listed in table 1-1. The Model 113BR indicates time on a standard clock face and provides 1-pps output pulses or ticks whose phase can be manually adjusted using a calibrated front-panel control. In operation,. the phasing control is adjusted to give time coincidence between the clock tick and a master timing tick. The time difference between two phase settings can be read directly from the phasing control and is used to compute system Oscillator frequency. 1-5. The Specification H02 113BR is a standard Model 113BR which has been tuned for sideral time operation, using an input frequency of approximately 100.27379091 kc. The tuned circuits which are affected include T1 through T8. 1-6. In this manual, the Model 113BR Frequency Divider and Clock is referred to as the “clock”. 1-7. The clock meets the performance requirements of military specification MIL-E-16400 and is suitable for mobile operation. Fail-safe operation is provided by using non-self-starting frequency dividers and a non-self-starting synchronous clock motor. The clock therefore either continues to operate without error or ceases operation completely in case of momentary failure of either the external Power supply or the external Frequency Standard.
These manuals are available for the above equipment:
|Hewlett Packard -- 113BR -- Service and User Manual|
|Manual Type:||Service and User Manual|
|Size:||20.43 Mbytes (21425846 Bytes)|
|Quality:||Scanned document, reading partly badly, partly not readable.|
|Upload date:||31 May 2015|
|Downloads:||460 since 31 May 2015|
SERIALS PREFIXED: 107 - This manual provides complete instructions on installation, operation, and maintenance of the Hewlett-Packard Model 113BR Frequency Divider and Clock (figure 1-1) and the Specification H02 113BR Frequency Divider and Clock. This manual applies directly to all instruments with serial numbers prefixed 107-. Serial number is stamped on rear of cabinet and on serial number tag on chassis inside cabinet.
|Start Free Download|
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.